Int J Oral Maxillofac Implants 32 (2017), No. 6 21. Nov. 2017
Int J Oral Maxillofac Implants 31 (2016), No. 5 (19.09.2016)
Page 1040-1048, doi:10.11607/jomi.4544, PubMed:27632258
Retention and Stability of Implant-Retained Mandibular Overdentures Using Different Types of Resilient Attachments: An In Vitro Study
ELsyad, Moustafa Abdou / Agha, Nora Nory / Habib, Ahmed Ali
Purpose: The aim of this study was to evaluate and compare the retention and stability of mandibular implant overdentures using different types of resilient attachments.
Materials and Methods: Two implant analogs were inserted in the canine areas of an acrylic edentulous mandibular model. A metal-reinforced experimental overdenture was constructed and connected to the implant analogs (on two occasions) using either resilient telescopic or Locator attachments. Locators were divided into three subgroups according to the degree of retention of the male nylon insert: Locator extra-light retention (blue insert), Locator light retention (pink insert), and Locator medium retention (transparent insert). Vertical and oblique (anterior, posterior, and lateral) dislodging forces were measured at the beginning of the study (initial retention) and after 540 cycles of denture insertion and removal (final retention).
Results: For all dislodging forces, Locator medium recorded the highest initial and final retention. Telescopic attachments recorded the lowest retention during vertical and anterior dislodging, and Locator extra-light recorded the lowest retention during lateral and posterior dislodging. For all types of Locator attachments, anterior dislodging recorded the highest initial and final retention, and lateral dislodging recorded the lowest retention. For the telescopic attachment, posterior dislodgment recorded the highest initial and final retention, and anterior dislodging recorded the lowest retention.
Conclusion: After repeated denture insertions and removals, the highest retention and stability were recorded with Locator medium followed by Locator light. The lowest retention was recorded with resilient telescopic attachment, and the lowest stability was recorded with Locator extra-light.
Keywords: attachment, implant overdenture, Locator, retention, telescopic